OE.D*.E** Oxygen Electrode
The Oxygen electrode consists of a PEEK body, platinum electrode, silver reference electrode and membrane holder. The platinum electrode is melted in glass. The membrane holder with electrolyte is attached by thread. Oxygen electrode (OE) is designed for measurements with thermostated cell TC4, TC5, TC6.
WCEc.W*.D* Working Classic Electrode ConicRead more
WCEc is a glass tube with a Pt or Au wire inside. Only circle glossy area of the metal wire is outside the glass tube. Standard connection is with a 2 mm banana connector. Connector or banana plug colour is red. Working classic electrode (WCEc) is designed for measurements with TC4, TC5, TC6, TC9 glass cell.
CEc Calomel Electrode ConicRead more
The calomel electrode (CEc) is a reference electrode based on the reaction between elemental mercury and mercury(I) chloride. The aqueous phase in contact with the mercury and the mercury(I) chloride (Hg2Cl2, “calomel”) is a saturated solution of potassium chloride in water. The electrode is linked via a porous frit to the solution in which the other electrode is immersed. This porous frit makes liquid junction. Calomel electrode (CEc) is designed for measurements with thermostated cell TC4, TC5, TC6, TC9.
RCEc.R* Reference Classic Electrode ConicRead more
RCEc is a glass hollow tube with silver or silver covered by silver chloride wire inside. The tube is hollow with the hole in the wall for inserting KCl solution. Hole is covered by a rubber band. Reference classic electrode (RCEc) is designed for measurements with glass cell TC4, TC5, TC6, TC9.
Standard connection is with a 2 mm banana connector.
Connector or banana plug colour is blue.